- First trials with 365nm (UV-A) were unsuccessful.
- Second trial with UV-CCFL spectrum - 254nm (UV-C) worked surprisingly well.
Final Test Results
The final test results summarized using the example of an IPU film:
According to the standard, the gray value difference may be 1%. In the most unfavorable point (with the greatest penetration) 0.5% was measured, machining with one minute at level 1 and 5 minutes at level 2.
Another test showed that the properties of the film (sensitivity) were not influenced by the erasure process.